Martin Luther King Jr. Day
Our office will be closed Monday, Jan 18th in observance of Martin Luther King Jr. Day. We will reopen at regular business hours on Tuesday, Jan 19th.
Development Of Wrist And Ankle Cut-points For Youth With The ActiGraph Accelerometer
- Presented on May 29, 2014
Purpose: The purpose of this study was to develop and validate physical activity (PA) cut-points for wrist (WR) and ankle (AK) accelerometer placements in youth.
Methods: : For cut-point development 84 girls and 97 boys (mean ± SD; age, 12.0 ± 1.5 yrs) completed 30 min of lying rest and two to seven structured activities (from a list of 25) for 7 min each (ranging from sedentary to vigorous intensity). Cutpoints were validated on 26 boys and 15 girls (age, 12.6 ± 0.8 yrs) who completed approximately 2 hrs of free-living activity. Activity data were collected using an ActiGraph GT3X or GT3X+, positioned on the dominant WR and non-dominant AK and were converted to 5 sec epochs for the vertical axis (VA) and the vector magnitude (VM). Simultaneously EE (METs) was measured using a Cosmed K4b2. Measured METs were calculated as measured activity VO2 divided by measured resting VO2. Receiver Operator Characteristic (ROC) curve analyses were used to develop WR and AK cut-points (VA and VM) for sedentary (SED; <1.5 METs), moderate (MPA; 3.0-5.99 METs), and vigorous (VPA; ≥ 6.0 METs) intensities. Repeated Measures ANOVAs were used to compare measured and predicted (ROC cut-points) time in each intensity category during free-living activity.
Results: WR cut-points for SED, MPA and VPA were: VA ≤ 105, ≥ 262, and ≥ 565 counts/5 sec; VM ≤ 275, ≥ 416, and ≥ 778 counts/5 sec. AK cut-points for SED, MPA and VPA were: VA ≤ 23, ≥ 90, and ≥ 202 counts/5 sec; VM ≤ 127, ≥ 309, and ≥ 515 counts/5 sec. All ROC analyses were similar for area under the curve (≥ 0.818), sensitivity (≥ 0.756), and specifi city (≥ 0.626). WR VA and VM cut-points signifi cantly underestimated measured time in light activity (LPA) by 13.0 – 18.8 min and overestimated VPA by 14.1 -15.9 min (P < 0.05), and were within 7.8 min of measured SED and MPA time (P > 0.05). AK VA and VM cut-points significantly underestimated measured time in LPA and MPA by 11.8 – 18.8 min and overestimated SED and VPA time by 12.0 – 16.5 min (P < 0.05). Individual errors tended to be lower for the AK (95% prediction interval (PI), ± 22.0 – 41.6 min) versus WR (95%PI, ± 30.0 – 49.9 min).
Conclusion: Compared to the Cosmed, the WR or AK cut-points have similar errors on a group basis. Additionally, the WR and AK errors are similar to published errors for ActiGraph cut-points used with a hip location.
Study supported by NIH grant R21HL093407