Our office will remain closed through Friday, September 18th as we continue to assess the damages caused by Hurricane Sally. ActiGraph team members are working remotely, however shipping delays should be expected at this time. We expect to resume regular business hours on Monday, September 21st. If you need immediate assistance, please contact us by email at email@example.com and we will respond as quickly as possible. Thank you for your continued support.
Mechanical and Free Living Comparisons of Four Generations of the Actigraph Activity Monitor
- Published on 2012
Background More studies include multiple generations of the Actigraph activity monitor. So far no studies have compared the output including the newest generation and investigated the impact on the output of the activity monitor when enabling the low frequency extension (LFE) option. The aims were to study the responses of four generations (AM7164, GT1M, GT3X and GT3X+) of the Actigraph activity monitor in a mechanical setup and a free living environment with and without enabling the LFE option.
Methods The monitors were oscillated in a mechanical setup using two radii in the frequency range 0.25-3.0 Hz. Following the mechanical study a convenience sample (N = 20) wore three monitors (one AM7164 and two GT3X) for 24 hours.
Results The AM7164 differed from the newer generations across frequencies (p < 0.05) in the mechanical setup. The AM7164 produced a higher output at the lower and at the highest intensities, whereas the output was lower at the middle intensities in the mid-range compared to the newer generations. The LFE option decreased the differences at the lower frequencies, but increased differences at the higher. In free living, the mean physical activity level (PA) of the GT3X was 18 counts per minute (CPM) (8%) lower compared to the AM7164 (p < 0.001). Time spent in sedentary intensity was 26.6 minutes (95% CI 15.6 to 35.3) higher when assessed by the GT3X compared to the AM7164 (p < 0.001). Time spend in light and vigorous PA were 23.3 minutes (95% CI 31.8 to 14.8) and 11.7 minutes (95% CI 2.8 to 0.7) lower when assessed by the GT3X compared to the AM7164 (p < 0.05). When enabling the LFE the differences in the sedentary and light PA intensity (<333 counts*10 sec-1) were attenuated (p > 0.05 for differences between generations) thus attenuated the difference in mean PA (p > 0.05) when the LFE option was enabled. However, it did not attenuate the difference in time spend in vigorous PA and it introduced a difference in time spend in moderate PA (+ 3.0 min (95% CI 0.4 to 5.6)) between the generations.
Conclusion We observed significant differences between the AM7164 and the newer Actigraph GT-generations (GT1M, GT3X and GT3X+) in a mechanical setup and in free-living. Enabling the LFE option attenuated the differences in mean PA completely, but induced a bias in the moderate PA intensities.
Link to Abstract: http://www.ijbnpa.org/content/9/1/113