ActiGraph has moved!
Please make a note of our new corporate address:70 North Baylen Street, Suite 400
Pensacola, FL 32502
Evaluation of the MyWellness Key Accelerometer
- Published on 06/11/2010
Objective To examine the concurrent validity of the Technogym MyWellness Key accelerometer against objective and subjective physical activity (PA) measures.
Design Randomised, cross-sectional design with two phases. The laboratory phase compared the MyWellness Key with the ActiGraph GT1M and the Yamax SW200 Digiwalker pedometer during graded treadmill walking, increasing speed each minute. The free-living phase compared the MyWellness Key with the ActiGraph, Digiwalker, Bouchard Activity cord (BAR) and Global Physical Activity Questionnaire (GPAQ) for seven continuous days. Data were analysed using Spearman rank-order correlation coefficients for all comparisons.
Setting Laboratory and free-living phases.
Participants Sixteen participants randomly stratified from 41 eligible respondents by sex (n=8 men; n=8 women) and PA levels (n=4 low, n=8 middle and n=4 high active).
Results There was a strong association between the MyWellness Key and the ActiGraph accelerometer during controlled graded treadmill walking (r=0.91, p<0.01) and in free-living settings (r=0.73-0.76 for light to vigorous PA, respectively, p<0.01). No associations were observed between the MyWellness Key and the BAR and GPAQ (p>0.05).
Conclusions The MyWellness Key has a high concurrent validity with the ActiGraph accelerometer to detect PA in both controlled laboratory and free-living settings.
Link to Abstract: http://bjsm.bmj.com/content/45/2/109.abstract
British Journal of Sports Medicine